AXS Vecta™ 74 - Size 0.074in by Stryker
This device is designed with more room for stent assisted coiling or balloon assisted coiling and the tip shape loss has been reduced by 28% from Synchro 2 Guidewires
Prepare your device:
Flush Excelsior microcatheter then insert synchro SELECT™ guidewire
Dual catheter technique:
Position the 1st catheter at the bottom of the aneurysm
Position the 2nd catheter at the neck the aneurysm
Carefully remove delivery wire and sheath together to prevent coil moving in the sheath
Deploy a loop of the 1st coil the the 2nd one.
Continue coiling simultaneously loop by loop
Ensure you see the T marker shape
Check with the 1 beep = contact then press button, 3 beep = detachment ok
Remove the 1st micro-Catheter from the bottom of the aneurysm. Repeat with the 2nd one
are necessary for technical reasons. Without them, this website may not function properly.
are necessary for specific functionality on the website. Without them, some features may be disabled.
allow us to analyse website use and to improve the visitor's experience.
allow us to personalise your experience and to send you relevant content and offers, on this website and other websites.